Is Trendelenburg standard on your chairs?
Only on certain models. Not everyone needs or wants the Trendelenburg feature, but you can get it as an option on our Classic Series recliners (54 Series, 56 Series, etc.) and our Ascent and Ascent II recliner lines. Trendelenburg is a standard feature on the Continuum, Passage, Concord, Elevate, and Alō.
Does my recliner have a quick-release seat?
Most Classic, Ascent, Ascent II and Continuum recliner models made after March 1999 have quick-release seats, as well as the new Alō recliner. Passage and Concord recliners do not.
How does the quick-release seat work?
It is a relatively simple operation. Once you do it, it quickly becomes second nature:
- Lock the front casters of the chair. Release the footrest and push the chair back into the full recline position.
- Locate the thumb screws under the front two corners of the chair seat. These screws will be bright yellow (or black on older models).
- Remove the screws and place the chair back into the full, upright position by first lifting the seat-back fully upright and then pushing the leg rest back down against the chair firmly until it latches.
- Standing in front of the chair, lift the seat front slightly and pull the seat toward you, pulling it out like a drawer. Once the seat is removed, you have full access to the inside of the chair for cleaning and maintenance.
- When you are finished, replace the seat by setting it back on the mechanism, making sure that the metal clip on each side of the seat bottom is under the flange on the mechanism. Slide the seat back into the chair. Make sure the bolts on the front corners of the seat are seated in the holes in the mechanism and replace the thumb screws to secure the seat.
Can I power wash my recliner?
No, never. We do not advise power washing the recliner, electronics or frame.
What is the weight capacity of my recliner?
Various models offer different levels of weight capacity:
- The Alō recliner comes in both a 300-pound and 500-pound patient weight capacity.
- Our standard Classic Series recliners have a 300-pound patient weight capacity, as do the Ascent, Ascent II and Passage recliners. They are marked with a red “capacity” label located on the frame at the back of the recliner.
- Bariatric recliners have a 500-pound patient weight capacity.
- The Continuum has a 400-pound patient weight capacity.
- The Concord has a 275-pound patient weight capacity.
- The Elevate Multi-Purpose chair has a 440-pound patient weight capacity.
Is my recliner under warranty?
All recliners are sold with a warranty. If you have questions or are unsure of the status of your warranty, please contact our Customer Care department at 800-998-5018 with the serial number of your recliner for details.
How do I specify a recliner by model number or "series" number?
To browse our products, start with the navigation bar at the top of the page. The Seating links will show you products based on types of chairs, while the Application links show products based on their clinical use. Once you have located the product that seems to best suit your needs, click “View Details” to take you to the product landing page for more information. For additional information, you can find brochures, videos and 3D models of our products on our Design Resources page.
In all cases, we encourage you to contact your local Champion representative or call us at the corporate office. We can help answer any questions and show you how to customize or accessorize your chairs, so you get the level of comfort and functionality you want and need.
How do I place an order?
Please contact your local Champion sales representative. If you do not know your rep, contact our corporate office at 800-998-5018 for assistance.
How do I arrange for repair or replacement of parts on my recliner?
First, locate your custom recliner’s serial number on the back of the chair, toward the bottom, on the left-hand side of the frame. After you have the serial number, please contact our Customer Care department at 800-998-5018.
Can I retrofit my recliner?
Some features and accessories can be retrofitted. Please locate the serial number on your chair first, then contact our Customer Care department at 800-998-5018 for details.
WHERE TO LOCATE YOUR SERIAL NUMBER:
How do I deploy the Trendelenburg option on a manual recliner?
For our Classic Series, Ascent, Ascent II or Alō recliners with manual recline that have Trendelenburg:
To deploy “Trend,” the patient must first be in a full recline position. Use the leg rest/recline release actuator located on the outside of either arm of the recliner. This will allow the leg rest to raise. If the patient is unable to assist you, manually raise the foot rest as you hold the release lever out to bring the patient’s feet up and achieve the “TV” position. From “TV” position, push the back of the recliner down using either the push handle located at the back or by simply applying pressure to the back of the recliner near the patient’s head.
Once the chair is fully reclined, pull and hold the Trend lever (located on the back of the recliner near the patient’s head) as you lower the chair back to the desired position. Again, do this while holding the Trend lever. (Please note that if you choose to use a “partial” Trend position instead of full Trend, you must always stop at a partial Trend on the way down. If the chair has been fully Trended, you must reset the Trend mechanism by returning the chair to the “TV” position while pulling and holding the Trend lever. Then go to the fully reclined position and stop at the desired position on the way down.)
To release the patient back into the full recline position from a Trended position, pull and hold the Trend lever as you raise the chair to the “TV” position. Release the Trend lever and push the chair back down to full recline with pressure on the back near the head or using the push handle at the back.
For the Continuum, Passage and Concord:
Use the foot release Trend, located at the bottom of the recliner. Make sure the patient is in full recline first, then depress the foot-release Trend lever. While the Trend lever is engaged, push the chair back down into Trend and release the foot lever. To release the patient back into the full recline position from Trend, simply pull the chair back up to the seated position.
How do I deploy the Trendelenburg option on a power recliner?
For our Classic Series, Ascent, Ascent II or Alō Power:
To deploy “Trend,” the patient must be in full recline position (if the patient is in other than full recline position, press the recline button on the handheld remote until the chair is in full recline). From there, pull and hold the Trend lever, which is located on the back of the recliner near the patient’s head. Lower the chair back to the desired position while holding the Trend lever by pressing on the chair back near the patient’s head area or using the push handle at the chair back. (Please note that if you choose to use a “partial” Trend position instead of full Trend, you may stop at any point by releasing the Trend lever. To return from Trend on a recliner with powered recline, pull and hold the Trend release as you lift the chair to the fully reclined position. Once you have reached the fully reclined position, release the Trend actuator. From this position, you may then engage the handheld remote to adjust the chair’s position in recline, or you may return to Trend if necessary.)
Where can I find the serial number of my chair?
The serial number is located at the back of the chair, near the back-left caster, on the metal frame. Look for the Champion label. The serial number is just below our name.
How can I determine the color code for my chair?
Simply contact us at Champion with the serial number of your recliner, and we can provide the color code for you.
How can I check the status of my order?
Please dial 800-998-5018.
Who is my local sales representative?
Champion has a nationwide network of sales reps to meet your needs. If you do not know who your local sales rep is, please contact us at 800-998-5018.
How do I obtain upholstery color swatches?
Your local Champion sales rep will have color cards on hand. If you do not know your rep, please contact us at 800-998-5018.
You can also view our online sample room and order specific graded vinyl samples by going to the Design Resources navigation at the top and then clicking on Fabrics & Finishes.
What are the electrical draw and specifications on the Champion heat, heat/massage and power recline options?
Contact us for more information at 800-998-5018.
Where are the chairs manufactured?
All Champion products are U.S. assembled in Elkhart, Indiana. The Elevate, distributed by Champion, is imported from Hungary.
What cleaning solutions are safe for my upholstery/tables?
Follow the manufacturer’s cleaning recommendations for your fabric or table options. The use of vinyl conditioners or protectants is not recommended. Some institutional cleaners or disinfectants may cause discoloration of the fabric, and embrittlement of the table.
Use of cleaners, other than those recommended by the manufacturer, is at the clinic’s own risk. Any type of cover you choose can be affected by long-term exposure to hair oils, body oils, lotions, personal care products, and other staining substances like shoe polish and ink. In addition, certain medications may produce a metabolite in the patient’s perspiration which can stain or discolor fabric. You may also want to research the materials that work best with your facility’s infection prevention protocol. If you have any questions, please refer to your Service Manual or call our Customer Service department at 800-998-5018.
For an overview of care and cleaning, watch our video:
How do I become a dealer or distributor of Champion products?
Please contact our corporate office at 800-998-5018.
I'm having difficulties viewing your website. What's wrong?
This site was designed with the latest web standards and security in mind and may not format correctly in some older browsers. While you can view this site with an older browser, you will have a better viewing experience by upgrading to a current browser.